Call Us Now! (03) 9874 email@example.com
Zero-drift and zero-shift in a strain gage Wheatstone Bridge circuit can arise from a number of sources, including the gage installation itself. The following checklist gives the recommended inspection steps immediately after the completion of lead wire attachment, but prior to the application of protective coatings.
After Leadwire Attachment
Most gage installations will exhibit none of these faults if the installation techniques outlined in Micro-Measurements instructional materials and training programs are followed. However, field applications made in harsh environments, or even laboratory installations with difficult gage locations on the test specimen, warrant special attention. If you have any questions concerning your particular applications, our Applications Engineering Department will be pleased to assist you in establishing a program for verifying the integrity of strain gage installations. They can be reached at (919) 365-3800 by telephone or by email at firstname.lastname@example.org.
Jim Johnson, Technical Sales Manager Micro-Measurements®, a Vishay Precision Group, Inc. (VPG) brand
To find out more, follow the link to the StrainBlogTM
To view the Micro-Measurements® products, follow the link
For a further discussion contact our Melbourne office by email or phone: +61 3 9874 5777
Since 1976, Applied Measurement
Australia Pty. Ltd. has taken pride in
providing innovative and effective
solutions for applications that involve
measurement of physical parameters.